Looking at some of the Power Outputs being claimed by various manufacturers of LED Red Light Therapy Units, it’s obvious that there are a lot of exaggerated and inaccurate claims being made.
Some Vendors publish their units Power Consumption numbers (instead of their Power Output in mW/cm2) which not only are misleading high, but totally irrelevant in terms of dosing.
Other Vendors typically measure their units Power Output at the Emitter surface (right up against the actual LED), to claim numbers about 3x higher than the actual output thru their optical system. Considering the lights are not used without their optics in place, those numbers would be useless to calculate dosing.
In an effort to avoid hype and provide accurate, reality based numbers to calculate dosing, we measured or Photopuncture Torch (on the HIGH mode) for actual Power Output (Energy Density) in mW/cm2 at various settings and distances with a Tenmars Power Meter. We also measured 3 other popular torches for comparison purposes. The Photopuncture Torch’s Power Output for Low mode is 33% of High mode.
Many large NIH Studies reflect a practical and effective range for Energy Densities of between 20 mW/cm2 and 310 mW/cm2.
Power Output/ Energy Density mW/cm2
Actual “real world” Energy Densities after exiting optics is about 30% of that at Emitter Surface. Use that # for dosing calculations
|At LED Emitter Surface||mW/cm2||Exiting Optics||mW/cm2|
|PTI Photopuncture Torch||1,000||PTI Photopuncture Torch||310|
|Photonic Health Pro-Gen II||n/a||Photonic Health Pro-Gen II||295|
|Aah Light||550||Aah Light||164|
Calculate Dose in J/cm2 for Photopuncture Torch at “0” Distance (against Treatment Area)
At “0” Distance, Photopuncture Torch Energy Density = 310 mW/cm2
310 mW/cm2 = .31 W Multiply by exposure in seconds for dose in J/cm2
Ex: .31 W x 15 secs = 4.65 J/cm2 Dose
Calculate Dosage for Photopuncture Torch at other distances from Treatment Area
at 1″ Distance = 2 1/4″ Beam Width Energy Density = 50 mW/cm2
2″ Distance = 3 1/2″ Beam Width Energy Density = 20 mW/cm2
50 mW/cm2 = .05 W Multiply by exposure in seconds for dose in J/cm2
Ex: .05 W x 90 secs = 4.5 J/cm2 Dose
.02 W x 200 secs = 4 J/cm2 Dose
The Photopuncture Torch also has a “Zoom” feature.
At the Maximum zoom setting you can use the following Energy Densities to calculate Dosage:
at 4″ Distance = 1 1/4″ Beam Width Energy Density = 66 mW/cm2
3″ Distance = 1 1/8″ Beam Width Energy Density = 67 mW/cm2
2″ Distance = 1″ Beam Width Energy Density = 70 mW/cm2
1″ Distance = 1″ Beam Width Energy Density = 75 mW/cm2
Example: 2″ Distance (Max Zoom) Energy Density = 70 mW/cm2 = .07W
.07 W x 60 secs = 4.2 J/cm2 Dose